Seamless Gas Cylinder Water Jacket Testing Machine introduction: The equipment is all made of stainless steel. First, connect the seamless gas cylinder filled with water with the sealing top cover, and then connect the sealing top cover and the sealing cylinder body through the locking jaw fixing connection to make the sealing cylinder completely sealed. Then the low-pressure water transmission two-way pipe works with the water supply pump, and the water is transported from the water inlet of the sealed cylinder to the sealed cylinder through the water inlet pipe of the inlet of the inlet of the water inlet of the water inlet. At this time, the first manual valve is open until the water in the sealing cylinder is filled, and the water flows out from the drain of the sealing cylinder through the outlet pipe and the drainage tee connection pipe to the outside, and then the first manual valve is closed so that the water overflowing in the sealing cylinder is no longer discharged to the outside. After the drainage tee connection pipe is discharged into the drainage elbow, and finally discharged into the measuring cup,until the water in the measuring cup is completely immersed in the output end of the drainage elbow, at this time, the second manual valve is closed, so that the water inlet of the sealing cylinder no longer enters the water, and the weight of the electronic scale is peeled and cleared to zero, and the drainage elbow will no longer come out of the water. Then through the high- pressure pump switch button, the high pressure pump works with the water tank,so that the high-pressure water passes through the auxiliary high-pressure water pipeline and the water module, and enters the seamless gas cylinder in the sealed cylinder through the high-pressure water pipe, and the water is injected into the seamless cylinder cylinder to pressurize, and the high pressure pump stops working after reaching the specified pressure. In the process of bottle body expansion, the overflow after expansion flows from the drain of the sealed cylinder through the drainage bend to the measuring cup, this part of the water volume is the volumetric deformation of the tested seamless gas cylinder, and then manually and slowly relieved the pressure through the manual pressure relief valve, the elastic deformation disappears after the bottle pressure is completely unloaded, the water in the measuring cup will flow back into the sealed cylinder during the pressure relief process, and the weight of the final water is the volume residual deformation of the tested seamless gas cylinder, and the ratio of the volume residual deformation to the total deformation is the volume residual deformation rate of the cylinder. So as to judge whether the product is qualified testing standards,the equipment meets the national standards.